Meiji MT6520 Binocular PCM Asbestos Microscope - NIOSH 7400, OSHA ID 160Meiji Techno
- Binocular head, 360° rotatable
- KHW10x/20mm eyepieces
- Walton & Beckett reticle
- Plan Semi Apochromat 4x, 10x
- Plan Semi Apochromat Positive Phase 40x Objectives
- 6V 30W Halogen illumination with variable control
- LIMITED LIFETIME WARRANTY
The MT6520 Newly Designed PCM (Phase Contrast) Compound Microscopes for Asbestos fiber counting applications. Meiji Techno manufactures these microscopes pursuant to NIOSH 7400 and OSHA ID 160 Reference methods. Each model comes with centering telescope, 1mm stage micrometer, GIF 546nm filters and built-in Koehler illuminator as standard equipment. The specifications table below shows the features of each model.
Asbestos Fiber Counting by Phase Contrast Microscopy
Methodology for PCM Analysis: NIOSH 7400 & OSHA ID160
Phase contrast microscopy is the method that is primarily used for estimating asbestos concentrations of airborne fibers. This method is quick and can be performed on-site for a rapid determination of concentrations of asbestos fibers in the air. Phase contrast microscopy does not positively differentiate between asbestos and other fibers. Positive identification of asbestos fiber must be performed using PLM (Polarized Light Microscopy) or Electron Microscopy (Method 7402)
Required Microscope Equipment
Positive phase contrast microscope with 10X Brightfield and 40X Phase objectives, GIF (Green Interference Filter), Walton & Beckett reticle Type-22G and a stage micrometer with 0.01mm divisions.
Required Test Slide
The HSE/NPL Mark II Phase Shift Test Slide checks or standardizes the visual detection limits of the phase contrast microscope. The HSE/NPL Test Slide consists of a conventional glass microscope slide with seven sets of parallel line pairs of decreasing widths. The microscope must clearly resolve line pairs 1 thru 3. Line pairs 4 and 5 must be at least partially visible. Line pairs 6 and 7 must be invisible. A microscope which fails to meet these requirements is either too low or too high in resolution and cannot be used for asbestos detection.
|Viewing Bodies||Binocular Siedentopf head, 30° inclination, graduated diopter on left eyetube (23.2mm I.D. eyetube) with interpupillary adjustments of 53mm to 75mm.|
|Eyepiece||KHW10X compensating eyepiece, Field No. 20, tube O.D. 23.2mm
KHW10X Focusing eyepiece with Walton & Beckett reticle, Field No 20 (Eyetube O.D. 23.2mm)
|Objectives||Infinity Corrected Plan Semi Apochromat 4x & 10x
Infinity Corrected Plan Semi Apochromat Positive Phase Contrast objective 40x
|Stage||Ceramic coated flat-top stage, 170mm x 140mm, with 80mm x 52mm X-Y movement and right hand drop-down coaxial controls|
|Condenser||Abbe condenser N.A. 1.25 with iris diaphragm with centerable phase annuli for 20x & 40x phase objectives, dovetail mount|
|Focusing||Coaxial coarse and fine focus adjustment with graduation, reading 2 microns per division with tension adjustment and safety auto focus stop.|
|Illumination||6V 30W Halogen Koehler illumination with Integrated Automatic Voltage Sensing Power Supply|
|Includes||Focusing Centering Telescope, 23.2mm O.D.
Green interference filter 546mm, 44.0mm diameter, unmounted
Blue clear filter, 44.0mm diameter, unmounted
Stage Micrometer-1mm divided into 100 units, 0.01mm(each division= 0.01mm)
|Warranty||LIMITED LIFETIME WARRANTY|
|Certification||CE and ISO 9001|
- 28.00 LBS
- 16.00 (in)
- 14.00 (in)
- 21.00 (in)
Calculate shipping costs at cart or checkout. Our representatives are also happy to assist you with a Shipping Quote.
If you are dissatisfied with your purchase, let us help you with a replacement, return or repair. Exclusions may apply.
For full details, view our Returns and Exchanges policy.